|Download Golay Detector Calibration Set GC-CS Datasheet (PDF, 170 KB)|
The Golay Detector operating principle is based on a physical process of heating the gas by radiation absorbed by a thin film. The efficiency of the process can vary according to the environment conditions (e.g. unstable temperature). It has an influence on the detector’s response. Long term use of the device may make the films (in the receiving chamber) stretch; it also affects the parameters and measuring results. Summarizing the above, voltage-power curve measured by the manufacturer cannot be completely relied upon for determining the calibration coefficient.
The best possible solution is calibrating Golay Detector at the user’s laboratory directly before the measurements. That is the purpose of the GC-CS Golay Detector calibration set.
The GC-CS Golay Detector calibration set is a suite of equipment for on-the-spot performance verification of detectors, evaluation of their stability and adjustment of calibration coefficients. The calibration set provides the environment for achieving maximum accuracy of absolute measurements with Golay Detector throughout its entire lifetime.
The key component of the set is a highly stable heat source reproducing radiation temperature of 100°C.
|Temperature reproduced by the radiation source, °C||100|
|Radiation source’s absolute confidence limits at 95% confidence, °C||±1|
|Temperature variability around set level, °C, no more than||±0.1|
|Radiation source temperature drift @ 15 minutes, °C, no more than||±0.1|
|Radiation source output aperture diameter, mm||60±2|
|Mains voltage, V||from 90 to 230|
|Mains frequency, Hz||50±0,5|
|Power demand, W, no more than||100|
|Ambient pressure range, mm Hg||760 ÷ 10-3|
|Operating and storage temperature, °C||15 ÷ 25|
|Humidity, %||40 ÷ 80|
|Radiation source dimensions, mm||260 x 180 x 120|
|Radiation source mass, kg||2,5|
For quotation and delivery please fill in our request form.
Optics is ready to be shipped!
Shipment/ Payment/ Warranty ...